USER FUNCTIONS Panel
Assign user-defined functions that will be called in response to acquisition events.
|Event Name||Select one of several events of interest.|
|User Function||The function to call in response to the event. Note that multiple events can be sent to the same function.|
|Arguments||A cell array with optional arguments to pass to your user function.|
|Enable||If this is checked, the user function will be called when the event is triggered.|
|Add or delete a user function. Delete operates on the currently selected row.|
|User functions can be stored in the CFG file or the USR file. Typically USR-defined user functions override those in CFG. The main table view is toggled to show either the CFG or USR set based on these buttons.|
|Saves the current set of user functions to the CFG or USR configuration.|
A user function has the format:
|source||The UserFunction Class. You can access the root SI object by accessing the source's hidden property `hSI`.|
|event||a MATLAB EventData Class|
|arguments||A cell array with optional parameters. Specified in the arguments column of the User Functions table.|
A Very Very Brief Example
|Called just before the start of an Acquisition Mode such as GRAB or LOOP. This may happen before any data is acquired.|
|Called just after the last frame at the end of an Acquisition Mode such as GRAB or LOOP.|
|Called just after the first frame gets acquired from an Acquisition.|
|Called just after the last frame gets acquired from an Acquisition.|
|Called just after an acquisition is aborted. Abort usually happens because the Abort button is pressed in the Main Controls.|
|Called just after the shutter is closed after acquiring a slice during a Slow Stack acquisition.|
|Called just after the user presses the FOCUS button in the Main Controls.|
Called just after the FOCUS is aborted. Similar to acqAbort but only in FOCUS mode.
|Called just after a frame is acquired.|
|Called when the digitizer experiences an overvoltage condition during an acquisition.|
|Fires when command to perform an on demand stim is received, before the trigger is sent.|
Fires when an on demand stim completes if multiple outputs are allowed; more stims can be externally triggered.
|Fires when an on demand stim completes if multiple outputs are not allowed; no more stims will occur until next on demand command.|
|Fires when an external on demand stimulus selection occurs.|
|Fires when a sequence stimulation begins.|
|Fires when a sequence element is complete and sequence advances to the next stim.|
|Fires when a single sequence is complete.|
Fires when the last sequence is complete.
|Fires when photostimulation module aborts.|