User Settings / Config
Displays name of currently loaded User Settings (USR) file. If none loaded, field is blank.
Launch dialog to select new USR file to load
Save current user settings to currently loaded USR file. (Use File >> Save User Settings As... from menubar to save current user settings to a new USR file)
Displays name of currently loaded Configuration (CFG) file. If none loaded, field is blank.
Launch dialog to select new CFG file to load.
Save current configuration to currently loaded Configuration (CFG) file. (Use File >> Save Configuration As... from menubar to save current configuration to a new CFG file).
ScanImage can be configured to collect a specified number of 2D image Frames, at each of a number of axial (Z) Slices. In Loop mode, these are repeated for a given number of Acqs (Acquisitions).
For Grab/Loop modes: number of Frames acquired/to-acquire at current Slice for the current Acq (if using LOOP)
For Grab/Loop modes: number of Slices acquired/to-acquire for the current Acq (if using LOOP)
For Fast Z Imaging, the number of Slices pertains to the current Volume (and Acq)
|For Loop mode: the number of times to repeat an Acq of the specified number of Frames & Slices|
For Loop mode: specifies time, in seconds, between the start of each Acq
Applies only when Acquisition Start Trigger is self-generated by ScanImage
For Focus/Grab modes: displays time, in seconds, since start of acquisition.
For Loop mode (self-triggered): displays countdown from specified Interval to start of next Acq
For Loop mode (external Start Triggering): displays time since last Acq was externally triggered.
|Sets stem of filename to be saved for next or ongoing GRAB/LOOP acquisition. The Acquisition # is appended to specified Basename to create the full filename, e.g.test0515_001.tif in example shown.|
Sets number to append to specified Basename to create the full filename.
Value is automatically incremented at end of each GRAB or LOOP mode Acq
Value can also be:
|Increment Acquisition # by 1.|
|Launch dialog to select save directory to which files are saved.|
|Determines whether disk logging functionality is enabled during GRAB/LOOP acquisitions.|
Specifies # of frames to average together before appending single (averaged) frame to current log file.
Value > 1 implies file averaging is enabled.
File averaging capability is separate from the display Rolling Average specified in Image Controls
File averaging implies data reduction: raw data is lost and file size is reduced
Acquisition Start Controls
ScanImage has 3 Acquisition Modes - FOCUS, GRAB, and LOOP.
The corresponding buttons start each mode, beginning or arming an acquisition, and the button changes to ABORT.
Pressing ABORT stops the mode and any ongoing acquisition.
FOCUS mode: Continuous acquisition without file-saving, until ABORT is pressed. Used for searching specimen, panning & focusing, etc.
|GRAB mode: Start (or arm) a single acquisition.|
LOOP mode: Start (or arm) a sequence of acquisitions (for up to the specified number of Acqs).
Globally enables/disable all external acquisition triggering signals for Grab/Loop modes (no effect on FOCUS mode).
Displays the current state of ScanImage. Can be one of the following:
The Zoom, Shift Fast/Slow, and Scan Angle Multiplier Fast/Slow controls specify the default ScanImage ScanField - a single two-dimensional area region of interest (ROI) for imaging.
If Multi-ROI (MROI) Imaging is enabled, these settings are ignored in favor of the configured mutiple 3D Regions of Interest/Stimulus Editor
|Specifies factor by which scan angular range is reduced, for both fast & slow dimensions. Adjusting zoom factor will set value to nearest allowed zoom increment.|
|Resets Zoom value to 1.0 - i.e. to full scan angular range supported by scanning hardware.|
|Specifies scan angle offset in specified scan dimension; this effectively shifts scan pattern laterally in the microscope's plane of focus. This value is in degrees. Adjustment is only allowed in slow scan dimension.|
|Displays the reduction factor for scan angle (lateral scan extent in microscope focal plane) in specified scan dimension, i.e. displays the aspect ratio of the scanned area.|
A Scan Angle Multiplier Slow value = 1 specifies a square area scan
A Scan Angle Multiplier Slow value ≠ 1 specifies a rectangular area scan
A Scan Angle Multiplier Slow value = 0 specifies a line scan
|Toggle button switching Scan Angle Multiplier Slow between 0 (line scan mode) and the current Base ROI value.|
Points scanner at center of field-of-view (following specified Shift values), changing button to PARK.
Scanner remains pointed to field center until PARK is pressed.
Used for scanner alignment
While scanner is pointed, beam power is adjusted to specified level and shutter is opened (if such hardware is configured in Machine Data File)
MROI & PhotoStim Launch
The Multiple Regions of Interest (MROI) and Photostimulation (Photostim) modes allow targeted laser scanning for imaging and stimulation within the scan volume, respectively
Fast Configuration Buttons
The Fast Configurations window, from File >> Settings >> Fast Configurations..., allows specified CFG files to be associated with buttons on Main Controls (and keyboard shortcuts) for one click (press) access to frequently-used configurations.
MAIN CONTROLS Panel - File Menu
|Load User Settings...||Displays name of currently loaded User Settings (USR) file. If none loaded, field is blank.|
|Save User Settings...||Save current user settings to currently loaded USR file.|
|Save User Settings As...||Launch dialog to select new USR file to save.|
|Load Configuration...||Displays name of currently loaded Configuration (CFG) file. If none loaded, field is blank.|
|Save Configuration...||Save current configuration to currently loaded Configuration (CFG) file.|
|Save Configuration As...||Launch dialog to save current configuration to a new CFG file).|
|Fast Configurations...||Launch the Fast Configurations window.|
|Set Save Path...||Launch dialog to select save directory to which files are saved. (Identical in function to the DIR... button in the main window.)|
|Save Last Acquisition As...||Launch dialog to save images from currently displayed channels to TIF format. Only actual image data is saved - the merge window and any tiled images are not saved.|
|Exit ScanImage||Exits ScanImage and exits Matlab.|
|Exit Matlab||Exits ScanImage and exits Matlab.|
MAIN CONTROLS Panel - Settings Menu
|Beams...||Launch the Beams Window.|
|Triggers...||Launch the Trigger Window.|
|Channels...||Launch the Channels Window.|
MAIN CONTROLS Panel - View Menu
|Raise Windows||Moves all currently open ScanImage windows to the foreground.|
|Show All Windows||Launch all ScanImage windows (including those that have not been opened) and moves them to the foreground|
|Alignment Controls||Launch the Alignment Controls window.|
|FastZ Controls||Launch the Fast Z Controls window.|
|Image Controls||Launch the Image Controls window.|
|Motor Controls||Launch the Motor Controls window.|
|Photostim Controls||Launch the Photostim Controls window.|
|Power Controls||Launch the Power Controls window.|
|Channel 1 Display||Launch the Channel 1 Display window.|
|Channel 2 Display||Launch the Channel 2 Display window.|
|Channel 3 Display||Launch the Channel 3 Display window.|
|Channel 4 Display||Launch the Channel 4 Display window.|
|Channel Merge Display||Launch the Channel Merge Display window.|